Do you need help?
Contact us with any question regarding your used Keysight equipment purchase!
FieldFox Handheld RF Analyzer / 100 kHz to 4 GHz
Cannot ship to:
Unit only ships within China.
Unit ships worldwide.
USD 12,708.45 Save 55%
|104||4 GHZ CABLE AND ANTENNA ANALYZER|
|010||VECTOR NETWORK ANALYZER TIME DOMAIN|
|030||REMOTE CONTROL CAPABILITY|
|208||USB POWER SENSOR MEASUREMENTS VERSUS FREQUENCY|
|230||4 GHZ SPECTRUM ANALYZER|
|236||INTERFERENCE ANALYZER AND SPECTROGRAM|
|302||USB POWER SENSOR SUPPORT|
|303||NETWORK ANALYSIS CAPABILITY|
|311||CHANNEL POWER METER|
Never Sacrifice Capability or Portability
Keysight's FieldFox portable analyzers are equipped to handle your toughest working environments during routine maintenance, in-depth troubleshooting, and anything in between. Choose the Keysight FieldFox configuration that best meets your needs.
- Standard model includes cable and antenna analyzer (CAT)
- Expand capabilities with optional T/R vector network analyzer, spectrum analyzer, built-in power meter, and more
- Save time by measuring return loss and distance-to-fault simultaneously
- Calibrate easily in the field with QuickCal
- Measure average and pulse power with a USB power sensor
- Use the industry's lightest all-in-one analyzer at only 2.8 kg (6.2 lb)
- CAT and VNA: 2 MHz to 4/6 GHz
- SA: 100 kHz to 4/6 GHz
Ready To Get Started?
What You Get
- Keysight N9912A-104 Network Analyzers
- Accessories only included if shown in the pictures
- Calibration certificate only included if shown in the pictures
What You Can Add
- KeysightCare Assured
- Fresh Calibration
- Calibration Plan
|Keysight New||Keysight Premium Used||Keysight Used|
|Savings||None||Up to 70%||Up to 90%|
|Condition||New||Like New||Working Condition ¹|
|Warranty||1 to 5 Years ²||Like New||90 Days Warranty|
Learn more about Keysight Premium Used Here
Customization is generally possible with our Keysight Premium Used equipment. The majority of our Keysight Premium Used equipment can be configured just the way you need it. Options can be added at additional cost.
Customizing Keysight Premium Used equipment might change the delivery and lead time of the item.
Some Keysight Used products cannot be customized. Use the contact option on the product page to check with the our eStore team.
Keysight Premium Used equipment typically comes with the same accessories and warranty as new products.
We currently only offer those accessories which are part of the listing. Other accessories can be purchased separately at extra cost with your purchase of a Keysight Used or Keysight Premium Used unit.
Please mention in the comment field which additional accessories or support you are interested in.
High-performance Vector Network Analyzers (VNA) are used to characterize and analyze RF and Microwave devices such as amplifiers, filters, power dividers/combiners, switches, antennas, and more. VNAs provide engineers with accurate device measurements including S-parameters, noise figures, return loss, and more.
A Vector Network Analyzer, a stimulus response testing equipment, is commonly used for research and development, design validation, failure analysis, and production testing by measuring the frequency response of component(s).
Yes, all VNAs are calibrated to ensure the highest quality results. The calibration process involves adjusting the instrument to eradicate the hardware systematic errors using traceable standards for accuracy and phase measurements.
How do you calibrate a Vector Network Analyzer?
The calibration process can be done manually or automatically, but in most cases, the manual method is used. The common methods of calibration include SOLT and TRL based on different parameters to remove systematic errors. Manual calibration involves selecting a known stimulus signal from a list of available signals within the VNA's software, then adjusting the amplitude and phase until the value corresponds to that from the known stimulus source. The network analyzer should be periodically calibrated with a reference calibrator to ensure the analyzer is measuring correctly. Periodic calibration guarantees the match with International standards.
Some VNAs have built-in error correction that may compensate for a variety of errors including thermal drift, time delay, source-load mismatch and optical path length variations.
A Vector Network Analyzer measures the amplitude as well as phase characteristics of one and two-port devices such as wireless components, microstrip lines, and branched structures.
A Network Analyzer measures the network parameters of two-port and arbitrary number of ports devices such as amplifiers, filters etc.
A Vector Network Analyzer measures complex component specifications at specific frequencies using a signal generator and a receiver.
A Spectrum Analyzer measures power across an entire frequency spectrum of the applied signal using a receiver only.
An oscilloscope measures external signals by representing voltage waveforms as a function of time, while a Vector Network Analyzer measures both the amplitude and phase monitoring the response of a network.
A VNA measures S11 by introducing a stimulus signal into the port of the device being tested, then measuring the reflected signal. The frequency at which this reflection occurs is then used to calculate S11.
S21 is used to measure how well a device rejects signals that are passed through the input port and reflected on the output port. The power transferred from the first port to the next can be determined using the S21 parameter.
Propagation and channel modeling is a technique for estimating the performance of wireless communications antennas and their effect on cell coverage.
The most significant parameters in propagation and channel modeling are path loss, Fresnel zone, path attenuation, and multipath fading.
A Smith chart is a graphic representation of a particular type of circuit operation. It shows the impedance, admittance, and conductance of an amplifier or other system component as it relates to frequency or voltage. In order to understand how a Smith chart operates, it is important to know the basics of transmission line theory.
Voltage standing wave ratio is the measure of the non-reactive voltage received at a point in space relative to cause by an AC source, thus representing the reflected power from the antenna source. It is typically specified as volts per meter (V/m) or millivolts per meter (mV/m).
A Vector Network Analyzer measures phase by introducing a stimulus signal into the port of the device being measured. The degree of phase shift is then noted. A separate calibration is used to determine the degree of phase shift when no stimulus signal is applied.
A VNA measures amplitude by introducing a stimulus signal into the port of the device, then measuring the strength. The magnitude of the reflected signal is then used to calculate amplitude.
Amplitude properties include the amplitude at a particular frequency, amplitude ranges, amplitude linearity, and amplitude ratio measurements.